The pulse magnetic therapy device uses a single pulse triangle wave to generate an impact magnetic field, which acts on the surface of the human spine, femur and other bones. Through electromagnetic biological effects, it activates the activity of bone cells, interferes with bone metabolism, increases bone density, and improves bone quality.
1. Magnetic induction intensity a) Round magnetic pad: 1st gear 20mT, 2nd gear 60mT, 3rd gear 100mT, 4th gear 150mT, 5th gear 200mT, tolerance ±30%; b) Square magnetic pad: 1st gear 20mT, 2nd gear 40mT, 3rd gear 60mT, 4th gear 80mT, 5th gear 100mT, tolerance ±30%; c) Magnetic field safety range: The magnetic field strength should be less than or equal to 0.5mT in the environment beyond 20cm from the magnetic pad.
2. Magnetic field output pulse waveform: unidirectional triangle wave.
3. Magnetic field output pulse frequency: 20 times/min～80 times/min, stepping 10 times/min, tolerance ±10%.
4. Magnetic field output pulse width: The first gear is 0.75ms, the second gear is 1.25ms, the third gear is 2ms, the fourth gear is 3.5ms, the fifth gear is 6.25ms, and the tolerance is ±15%.
|Output Pulse Waveform
||Unidirection Triangular Wave
|Output Pulse Frequency
|Output Pulse Width
||0.75ms, 1.25ms, 2ms, 3.5ms, 6.25ms
|Round Magnetic Pad
||20mT, 60mT, 100mT, 150mT, 200mT
|Square Magnetic Pad
||20mT, 40mT, 60mT, 80mT, 100mT
It is suitable for the adjuvant treatment of pain caused by soft tissue sprains.
Patients with acute purulent inflammation, severe heart, liver, and kidney disease, bleeding or bleeding tendency, pacemaker carriers, tumors, low white blood cells, local metal foreign bodies, and magnetic allergies.